Holmarc’s Laser Raman Spectrometer (Model No:HO-EDS-06) is a useful instrument for the identification of wide range of substances in physics and chemistry laboratories. The procedure followed in this equipment is illuminating a sample with DPSS green laser and using a monochromator to examine the light scattered by the sample. It is a straight forward, non destructive technique requiring no sample preparation.
This spectrometer is designed for recording raman emissions from solids as well as liquids when a laser beam is passed through the sample. The apparatus suits Chemistry as well as Physics labs for characterizing materials by recording raman emissions. The set up consists of 40mW DPSS laser (532 nm), collection optics, sample mount, stages, monochromator and detector.
The monochromator is PC controlled and motorized. The experiment is conducted by scanning the emission spectrum by monochromator and recording the intensity of each interested wavelength from the detector. The readings are plotted on a graph. Monochromator used in the system has high resolution and low stray light.
When a light beam emitted by the laser device passes trough an external optic path and irradiates the sample, the scattered light enters the monochromator. When the grating in the monochromator is rotated, light signal passes through a slit and falls on a highly sensitive detector (PMT). The detector output is sent to the computer for further processing.
|Laser source||:||532 nm 40 mW DPSS Laser|
|Wavelength Range||:||200 ~ 800nm (Monochromator)|
|Wavelength Accuracy||:||≤ 0.4 nm|
|Wavelength Repeatability||:||≤ 0.2 nm|
|Reciprocal of Linear Dispersion||:||2.7 mm|
|Half-Width of Spectral line||:||≤ 0.2 nm @ 586 nm|
|Relative Aperture Ratio||:||D / F 1 / 5.5|
|Optical Grating||:||1200 l / mm|
|Slit Width||:||0 ~ 2mm Continuously Adjustable|
|Computer-controlled, user-friendly interface|
|Capable of automatic recording of Raman spectra|
|Wavelength Selection Method|
|DPSS laser is used as light source|
|Both solid and liquid samples can be analyzed|
|The system is assembled as a standalone unit with a footprint of 800 mm x 600mm|
|High sensitive PMT is used as detector|
|PMT protection circuit|
|Both manual and automatic recording of data possible|
Documents / Drawings